Long-Stroke Tensile Testing System for Vulcanized Rubber
Capable of Measuring Even Specimens with a Large Break Strain
A tensile test method for vulcanized rubber is stipulated in JIS K6251 (Rubber,vulcanized or thermoplastic-Determination of tensile stress-strain properties). One of the properties of vulcanized rubber is significant elongation. This long-stroke tensile testing system for vulcanized rubber has a long stroke that is required for calculating these tensile characteristics of rubber, and accommodates dumbbell specimens Nos.1 through 6 stipulated in JIS K6251.
This system is based on the Shimadzu Autograph AG Series, and consists of a "pantograph type grip" having outstanding operability, "SES-1000 Extensometer for Soft Materials" compatible with measurement of high ductility elongation, and the AG dedicated software "TRAPEZIUM2" for conducting data processing from the test force-displacement curve in real time.
This system enables you to efficiently obtain the tensile characteristics of vulcanized rubber.
|Load capacity||1 kN|
|Grip||1 kN pantograph grip type 2|
|Initial grip space||Dumbbell specimen No.1: 90 mm
Dumbbell specimen Nos.2, 3, 4: 70 mm
Dumbbell specimen No.5: 85 mm
Dumbbell specimen No.6: 40 mm
|Tensile stroke||Dumbbell specimen No.1: 655 mm
Dumbbell specimen Nos.2, 3, 4: 675 mm
Dumbbell specimen No.5: 660 mm
Dumbbell specimen No.6: 705 mm
|Displacement gauge length||20, 25, 40 mm|
|Displacement gauge measurement range||100, 200, 1000 % at each gauge length|
Vulcanized rubber, Rubber
* Appearance and specifications are subject to change without notice.