XSeeker 8000 - Specs
Microfocus X-Ray CT System
|X-Ray Generator Output||160 kV 1.2 mA|
|X-Ray Detector||Flat panel detector|
|Max. Input Resolution||Approx. 5.6 million pixels|
|Max. Sample Size/Weight||SRD*1 = 475 mm||ϕ300 mm×H 320 mm,10 kg (Including sample setting jig)|
|SRD = 550 mm||φ150 mm×H 320 mm,10 kg (Including sample setting jig)|
|Max. CT-Scan Region||SRD = 475 mm||φ85 mm×H 70 mm|
|SRD = 550 mm||φ100 mm×H 80 mm|
|Spatial Resolution||100 µm (SRD = 550 mm)|
|CT Data Acquisition Time||12 sec or 5 min|
|Main Unit||Single phase 100 to 240 V AC ± 10 %, 50/60 Hz, 1.0 kVA|
|Control Computer||Single phase 100 to 240 V AC ± 10 %, 50/60 Hz, 1.0 kVA|
|Grounding||D type grounding (Grounding resistance of 100 max.)|
|Operating Environment Conditions||Ambient temperature: 10 to 30 °C; Ambient humidity: RH of 45 to 80 % max. (No condensation)|
|External Leakage Dose||1Sv/h or less.|
*1 SRD: The source-to-rotation center distance (SRD) is the distance from the X-ray source to the rotation center of the sample.
XSeeker 8000 Image and Layout (Units: mm)
Recommended Installation Space: W1950 × D1000 mm
Note: Leave Leave at least 350 mm of space behind the CT system table and 150mm of space left side in order to permit access for maintenance work.
XSeeker and the Analytical Intelligence logo are trademarks of Shimadzu Corporation or its affiliated companies in Japan and/or other countries.
VGSTUDIO MAX is a trademark of Volume Graphics GmbH.