Nexera Mikros - Features
Supporting Micro Flowrate Range Liquid Chromatography Mass Spectrometry System
Incomparable Microflow Ruggedness
Realize the Benefits of Microflow While Enjoying the Ruggedness and Reliability of HPLC
The Nexera Mikros system delivers the improvements you expect from a low-flow system with the ruggedness of HPLC. A plasma blank was protein precipitated, diluted (3x), and injected 1,500 times consecutively. During the consecutive injections, QC sample (40 ppt of nortriptyline) was injected once every 50 injections of plasma blank. Peak areas and retention times of QC sample are plotted below. With 4.38% RSD area repeatability and 0.25% RSD retention time stability, extremely stable operation was demonstrated, without loss of peak shape common in high-throughput analyses of complex matrices.
*Results obtained under our application conditions.
Download the latest brochure.
Nexera Mikros - Microflow Liquid Chromatography Mass Spectrometry System [ PDF / 1.73MB ]