Flexural Test Methods for Plastics: ASTM D790
Resin materials (plastics) are found in a wide variety of items used on a daily basis. Recently, plastics have started to be used as structural materials in transportation equipment, such as automobiles and aircraft, due to their strength and light-weight nature. In these applications, it is important to understand the mechanical strength properties of these plastics.
ASTM D790 specifies methods for determining the flexural properties (bending properties) of reinforced and unreinforced plastics and electrical insulation materials.
The standard includes some unique requirements that differ from the equivalent ISO and JIS standards, ISO 178 and JIS K 7171. For more information about ISO 178 and JIS K 7171,please click here.
ASTM D790 requires a test force accuracy that meets ASTM E4 Class 1, accuracy within ±1% of the indicated test force, which is equivalent to ISO 7500-1 Class 1 and JIS B 7721 Class 1.
During testing, ASTM D790 requires the use of a deflectometer to measure flexural displacements. Shimadzu's ASTM D790 compliant deflectometers are compatible with AG-X plus series, AGS-X series, and EZ-X series test frames. With the addition of Trapezium X testing software, a comprehensive flexural testing package is created.
In addition to the specimen deflection during flexural testing, crosshead displacement values can include the penetration of the punch and supports into the specimen and the deflection of the testing machine itself. ASTM D790 specifies deflection during testing to not exceed 1% of the specimen strain.
The radius of the punch and support points for the 3-point bend test fixture is specified in ASTM D790.
Different radius values for punches and supports are required for different thickness specimens....
News / Events
Shimadzu's New HMV-G30 / HMV-G31Series Micro Vickers Hardness Testers Include New Data Protection Functions and a Model Equipped with a Color Camera
Shimadzu micro Vickers hardness testers are equipped with excellent automatic reading functions and visually straightforward, convenient software. In addition, an extensive lineup of options is available to accommodate a variety of needs.
AUTOGRAPH AGX-V Series Video are now available.
Shimadzu's New AGX-V Series Autograph Universal Testing Machines Achieve World's Highest Performance Levels
This is the 100th anniversary of manufacturing Shimadzu testing machines.
The AGX series is the culmination of the further evolution of the AUTOGRAPH based on our wealth of experience and achievements
Evaluation of Dynamic Deformation Behavior of LPSO Type Magnesium Alloy by AE Method and High Speed Camera
Automobile Evaluation Instruments Brochure is now available.
Amino acid composition analysis has traditionally been conducted for protein quantitation and peptide structure prediction. Its use has also extended in recent years to the quantitation of such functional components as branched chain amino acids (BCAA). Previously, in Application News No. L432 and L437, we introduced examples of analysis of amino acids subjected to fluorescence derivatization using o-phthalaldehyde (OPA) using the SIL-30AC. Here, we introduce an example of fast analysis of amino acids in proteins.